please how to use the Isolated UF
What type of machines do you have?
These are the steps
Set the UF Volume
Set the time
Press Start/Stop UF button (until the light remains on)
When UF phase has finished, time remaining displays 0.00, an alarm appears and 'Time' button flashes
Press the isolated UF button for 3 seconds until light goes off
Set time for dialysis treatment
Set machine for dialysis treatment as required. You must add the isolated UF volume onto the UF volume for dialysis treatment.
Hope these help
|Powered by Social Strata|