Have a machine that's failes actuator venous high and low and venous high low soft what do I need to look for to fix the problem. I'am need to Biomed all the help will help me out.
blow out and reseat all your boards in card cage
let machine come up to cond/temp insure that your temp is set to 37, insure that your art/venous pressures are a zero with the port open, often times something will not pass and cause alot of other alarms to fail watch the test
disclaimer my views and no one else
|Powered by Social Strata|